FLASHTM Ostial System is designed to stabilize, post-dilate and conform the stent to the ostium during ostial post-stent dilation and angioplasty. The proprietary dual balloon design enables the physician to achieve maximal stent wall apposition after post-dilation and result in treating Ostial lesions. The FLASHTM is available for both coronary and peripheral indications.
Procedural steps: stent post dilatation
Distal always inside stent
Middle at ostium
Proximal outside guide and in aorta
Inflate Distal balloon
For post stent dilation.
Inflate Proximal balloon
For better stability and apposition of the stent to the ostium
(Use 1ml syringe for FLASHTM and 0.4 ml syringe for FLASHTMMINI)
Indications For Use: The FLASH™ Ostial System is indicated for balloon dilatation of the stenotic portion of a coronary artery or bypass graft for the purpose of improving myocardial perfusion. The FLASH™ Ostial System is also indicated for the post delivery expansion of balloon expandable stents within the coronary vasculature. Note: The FLASH Ostial System was tested on the bench with Boston Scientific VeriFLEXTM (a.k.a Liberté) balloon expandable stents. Consideration should be taken when this device is used with different manufacturers’ stents due to differences in stent design. All stents should be deployed in accordance with manufacturers’ indications and instruction for use.
Contraindications: Unprotected left main coronary artery. Coronary artery spasm in the absence of a significant stenosis.
Warnings (all sizes): Contents are supplied STERILE using radiation (e-beam) and are non-pyrogenic. Do not use if sterile barrier is opened or damaged. This device is intended for single use only. Do not reuse, reprocess or re-sterilize. Balloon and/or catheter integrity may be compromised by reprocessing or re-sterilization and could lead to serious patient injury. When the catheter is exposed to the vascular system, it should be manipulated while under high-quality fluoroscopic observation. Do not advance or retract the catheter unless the balloons are fully deflated under vacuum. If resistance is felt during manipulation, determine the cause of the resistance before proceeding. Applying excessive pull force to the catheter can result in tip breakage or balloon separation. To reduce the potential for vessel damage, the inflated diameter of the Angioplasty Balloon should approximate the diameter of the vessel or graft just proximal and distal to the stenosis. Do not exceed the rated burst pressure or maximum inflation volume recommended per the compliance table on the product labeling. The rated burst pressure is based on the results of in vitro testing. At least 99.9 percent of the balloons, (with a 95 percent confidence) will not burst at or below their rated burst pressure. To prevent over pressurization, use of a pressure monitoring device is recommended for Angioplasty Balloon inflation. To reduce the potential for air embolus into the vessel, use only the recommended balloon inflation medium (50% Contrast / 50%Sterile Saline). Never use air or other gaseous medium to inflate the balloon. PTCA in patients who are not acceptable candidates for coronary artery bypass graft surgery warrants careful consideration, including possible hemodynamic support during PTCA, as treatment of this patient population carries special risk. The FLASH™ Ostial System is not cleared for expanding balloon expandable stents within the neurovasculature.
Warnings (FLASH™Mini with Pressure Relief Syringe): If the pressure relief bladder at the distal end of the 1.0cc syringe barrel is inflated for more than 15 seconds, deflate both the Proximal Balloon and the Angioplasty Balloon and then reposition the catheter. Do Not switch to a different inflation device. Note: The pressure relief bladder is designed to inflate when the catheter has been incorrectly positioned. When repositioning the catheter, ensure that the marker bands are correctly positioned as indicated in the Angioplasty / Post-Delivery Stent Dilatation section of this IFU. Do not pinch or manipulate the pressure relief bladder when it is inflated. This could potentially result in vessel damage if the catheter has not been properly positioned.
Potential/Adverse Events: The complications that may result from a balloon dilatation procedure include: death, acute myocardial infarction, acute vessel closure, total occlusion of the coronary artery or bypass graft, coronary vessel dissection, perforation, rupture or injury, restenosis of the dilated vessel, hemorrhage or hematoma, unstable angina, arrhythmias, including ventricular fibrillation, drug reactions, allergic reaction to contrast medium, hypotension, hypertension, infection, coronary artery spasm, arteriovenous fistula, stroke, air embolism and embolization or fragmentation of thrombotic or atherosclerotic material.
Precautions: See full IFU for a list of Precautions.
Indications For Use: The FLASH™ Ostial System is indicated for Percutaneous Transluminal Angioplasty in the peripheral vasculature at aorto-ostial locations, including iliac, renal and carotid arteries. This device is also indicated for post-dilatation of balloon expandable stents in the peripheral vasculature.Note: The Flash Ostial System was tested on the bench with the Boston Scientific Express® SD balloon expandable stent. Consideration should be taken when this device is used with different manufacturers’ stents due to differences in stent design. All stents should be deployed in accordance with manufacturers’ indications and instruction for use.
Contraindications: None known.
Warnings: Contents are supplied STERILE using radiation (e-beam) and non-pyrogenic. Do not use if sterile barrier is opened or damaged. This device is intended for single use only. Do not reuse, reprocess or re-sterilize. Balloon and/or catheter integrity may be compromised by reprocessing or re-sterilization that could lead to serious patient injury. Use the catheter prior to the "Use By" date specified on the package. When the catheter is exposed to the vascular system, it should be manipulated while under high-quality fluoroscopic observation. Do not advance or retract the catheter unless the balloon is fully deflated under vacuum. If resistance is felt during manipulation, determine the cause of the resistance before proceeding. Applying excessive pull force to the catheter can result in tip breakage or balloon separation. Do not exceed 2.0 lbs when retracting the device into the guide catheter. To reduce the potential for vessel damage, the inflated diameter and length of the balloon should approximate the diameter and length of the vessel or graft just proximal and distal to the stenosis. Do not exceed the maximum burst pressure or maximum inflation volume recommended per the compliance table on the product labeling. To prevent over pressurization, use of a pressure monitoring device is recommended for angioplasty balloon inflation. Use the recommended balloon inflation medium (50% Contrast / 50% Sterile Saline). Never use air or other gaseous medium to inflate the balloon.
Potential/Adverse Events: The complications that may result from a peripheral balloon dilatation procedure include: Additional intervention, Allergic reaction to drugs or contrast medium, Embolization, Hematoma, Hemorrhage, Inflammation, Ischemia, Sepsis/Infection, Thrombosis, Vascular trauma (vessel dissection, spasm, etc.).
Precautions: See full IFU for a list of Precautions.
Legal Notices / Terms and Conditions
This website is maintained by Ostial Corp., solely for the purpose of providing general information about Ostial Corp., its business and products. The information on this website is provided for informational purposes only and is not intended or recommended as a substitute for professional medical advice. Always seek the advice of your physician or other qualified health care provider regarding any medical condition or treatment. We may also include certain information, reference guides and databases intended for use by licensed medical professionals. These tools are not intended to give professional medical advice. Physicians and other health care providers should always exercise their own clinical judgment for any given situation.
IF YOU THINK YOU HAVE AN EMERGENCY CONTACT YOUR LOCAL EMERGENCY CARE PROFESSIONAL OR FACILITY IMMEDIATELY.
Terms and Conditions of Use:
All of the content of this website (e.g., all information, text, photographs, illustrations, graphics, sound, video, meta tags, computer code, etc.) is owned by or licensed to Ostial Corp. and is protected under the copyright laws; as well as other laws, of the United States and other jurisdictions. Users of this website are permitted to download content of this website only for personal non-commercial purposes or in furtherance of the user’s dealings with Ostial Corp. No other downloading, copying, reproduction, modification, publication, broadcast or other use of any content of this website is authorized or permitted.
No representation, warranty or assurance of any kind is made regarding how current the information is on this website, or the accuracy or completeness of any content of this website. The owners of this site shall not be liable for any damages or injury resulting from anyone’s access to, or inability to access, this website, or from anyone’s reliance on or use of any content of this website.
This website may contain links or references to other websites. The owners of this website do not monitor or control the content of those other websites, have no responsibility for the content of such other websites and shall not be liable for any damages or injury arising from the content or use of such other websites. All links are provided merely as a convenience to the users of this website.
The trademarks, service marks, trade names, trade dress, and products in this website are protected in the United States and Internationally. No use of any of these may be made without the prior, written authorization of Ostial Corp., except to identify the products or services of Ostial Corp.
Collection of Information
As users navigate this website certain anonymous information may be passively collected (e.g., gathered without the user actively providing the information). Such collection of information may occur using various technologies, such as cookies, tags or web beacons, and / or navigational data collection (log files, server logs, clickstream). Some or all Internet browsers may automatically transmit to this website certain anonymous information, such as the URL of the website the user came from, the user’s Internet Protocol (IP) address, and the browser version being operated by the user’s computer. Some or all Internet browsers may allow users to be notified when a cookie is sent or to block cookies altogether, but certain features of this website might not work, or may be less than optimal, without cookies.
This website may use and combine such passively collected information to provide better service to users of this website, customize or improve the website based on users’ preferences, compile and analyze statistics and trends, administer and improve the website or for other such purposes as Ostial Corp. may deem appropriate. Ostial Corp. makes no representations or warranties regarding the use of such information.
Registration and Providing Information
This website may contain interactive areas or forms wherein patients, physicians, and others may register or provide their personal information, request further information, etc. By registering or providing personal information or information of any kind, users of this website represent that the information provided is true, accurate, current, and complete.
The sender of any communication or information to this website or to Ostial Corp., or its affiliates via this website, shall bear all responsibility for the content, legality, truthfulness, and accuracy of such communication or information.
Ostial Corp. will not share any personal information you provide with any third party absent your request to do so or your permission.
Patients & Physicians
United States laws, as well as the laws of other jurisdictions, restrict the sale and use of devices and products described in this website to sale and use by or on the order of a licensed physician.
Comments and Questions
Any comments or questions should be directed to Ostial Corp. We welcome your comments and feedback, but please understand that any information submitted is considered non-confidential and non-proprietary. You hereby agree that with respect to any such information provided to Ostial Corp. through this website that Ostial Corp. shall have a non-exclusive, unrestricted royalty-free license to use such information for any purpose.
ALL CONTENT IS PROVIDED TO YOU ON AN AS IS BASIS. OSTIAL CORP. MAKES NO REPRESENTATIONS OR WARRANTIES OF ANY KIND, EITHER EXPRESSED OR IMPLIED, REGARDING THE CONTENT OF THIS WEBSITE.